This article has Open Peer Review reports available.
A breath of fresh air: a quality-improvement study comparing an air-circulating technique versus conventional technique to prevent nasogastric tube dysfunction
© Bani Hani et al. 2015
Received: 5 November 2014
Accepted: 9 November 2015
Published: 27 November 2015
Nasogastric tubes are an important component of care in patients with gastrointestinal obstructions. However, they are prone to malfunction despite conventional flushing techniques, with potentially severe consequences. There is no widely accepted, gold-standard way to ensure that a nasogastric tube succeeds in maintaining an empty stomach following flushing.
We have developed a flushing technique to better ensure successful tube function. We compared this technique to conventional flushing both in vitro (using a plastic stomach model) and in vivo (in a pig model), and we provide a didactic video.
The mean gastric residual volume following our novel flushing technique is nearly 25-fold lower than following conventional flushing (13 mL vs. 330 mL).
Our simple technique is more effective than conventional techniques in maintaining nasogastric tube function and therefore should prevent dangerous vomiting and aspiration pneumonia better than conventional techniques.
KeywordsNGT Nasogastric tube Aspiration Pneumonia Patient safety
The stomach, however, is poorly suited to empty via an NGT. While larger (14- or 18-F) tubes are more effective at gastric emptying than smaller tubes (10-F) , even 16- or 18-F NGTs may fail to empty the stomach reliably and prevent vomiting. This may be due to the slack and mobile mucosal folds that become suctioned onto tube ports, evidenced by suction lesions at endoscopy . Conventional NGT maintenance includes simple flushing, which will push this mucosa away, but suction easily pulls it back into the orifices. Although conventional flushing will help avoid mucous plugging, it does not address the suctioning of gastric mucosa back into the NGT orifices, which may creates a one-way valve, such that fluid may pass into the stomach during conventional flushing, but not out of the stomach when suction is applied. This is why simply replacing the NGT fails to solve the problem, because the problem is not with the tube, it is with the stomach. The NGT must be therefore be assessed for patency and function in both directions, not only antegrade into the stomach, but retrograde out of the stomach.
We hypothesized that a flushing technique that ensures bidirectional patency and function of the NGT would more effectively ensure an empty stomach. The basis of the new technique is the fact that, while suction is applied to the main port, air injected into the side port should return immediately out the suction port of an appropriately placed NGT if and only if the stomach is empty. Accordingly, the observation of air circulating out the suction tubing immediately after being injected in the blue side port is a key indicator of bidirectional patency and function. By contrast, air injected into the side port would not be expected to suction out of the main port if the end of the NGT is submerged in a pool of gastric fluid, because the air would simply bubble to the top of the pool (Figs. 1, 4 and 5). We have designed a new flushing technique that relies on this visualization of flushed air to circulate immediately out the suction tubing.
When an extensive literature search failed to locate such a technique, we proceeded to assess our new flushing technique, comparing it to the conventional technique taught in nursing textbooks. Since vomiting is well recognized to occur in patients even following “successful” conventional NGT flushing, a better way to ensure an empty stomach is essential.
The lack of guiding literature on the best way to ensure proper function of an NGT for gastric decompression is well recognized [6, 7]. Current evidence-based recommendations to prevent aspiration pneumonia include maintaining the head of the bed at 30–45 degrees and ensuring proper NGT placement [8–10], or entirely redesigning the NGT [11, 12], but there is little guidance regarding assessing the NGT as the suction device in current widespread use. Evidence-based reviews suggest checking gastric residual volume (GRV) (defined as the volume left in the stomach after flushing an NGT) during gastric feeding [10, 13, 14] but do not specifically address how best to assess GRV, nor how to ensure that the measured GRV is accurate.
Therefore, a comprehensive literature search was conducted using PubMed and the inclusive keywords “nasogastric tube” with the following terms: flush (14), irrigation (74), suction (137), maintenance (61), care (1134), function (1080); aspiration pneumonia (186), residual (62), small-bowel obstruction (92), sump (15). The resultant list of 2855 citations contained 2214 unique citations. An additional search of simply “nasogastic tube” was performed to avoid missing any relevant citations and returned 3970 results, which were limited to adults, to the English language, to human studies, and to studies containing “nasogastric tube” in the title, leaving 325 citations. These 325 titles were independently scanned by two authors (MHB and SCC) for relevance regarding the care and maintenance of the NGT, the checking of gastric residuals, and the prevention of aspiration pneumonia. Relevant papers were then reviewed and manual cross-referencing was performed using each article’s reference list, to identify all published data about NGT function and care, which revealed no additional relevant studies.
In vitro experiments
An in vitro model of the stomach was designed using floppy plastic bags (grocery bags) modeling the floppy gastric mucosa, placed within 500-mL rigid plastic canisters modeling the more rigid gastric wall. This in-vitro model of the stomach was filled with warm tap water and an NGT was placed into the water with the end of the NGT in the most dependent portion.
Suction was applied to the NGT at -40 mm Hg, and the water was allowed to aspirate until a point at which either the model stomach was empty or fluid ceased to aspirate out (and remained ceased for 5 min), suggesting a blockage. The aspiration process was closely inspected for signs of blockage. If the plastic bag was seen to suction into the NGT holes then this was recorded as the reason for blockage (this occurred every time). If this was not seen, then the NGT was disconnected from suction tubing to ensure that there was still suction present (this latter case never occurred). Once this point was reached, the two flushing techniques (the conventional versus the new, air-circulating techniques) were employed as described below. Suction was then reapplied and the water was allowed to aspirate again, until a point at which either the modeled stomach was empty or fluid ceased to aspirate out (and remained ceased for 5 min), suggesting a recurrent blockage. The trial was then terminated and the volume of fluid remaining was measured as the GRV. Experiments were done in triplicate.
To validate that the plastic bag would indeed model the musoca of the human stomach, this model was compared to a similar set-up with the same plastic canister but without the plastic bag. The same volume of water, placement of NGT, flushing and aspiration were used.
In vivo experiments
With approval of the Johns Hopkins Animal Care and Use Committee and in accordance with the Animal Welfare Act, healthy swine (Sus domesticus) underwent laparoscopy followed by laparotomy during a Covidien-sponsored educational lab at the Johns Hopkins Minimally Invasive Surgical Training and Innovation Center, then were euthanized as planned. An NGT was immediately inserted through the mouth and into the stomach, where its position was confirmed by palpation through the stomach wall. Pigs whose stomachs were not empty, or were excessively distorted by the educational lab procedures, were excluded. A single pig was used, approximately 55 lbs, 4 months old, and female.
As an in vivo model of the human stomach, the pig stomachs were filled by instilling warm tap water via the NGT and suction was applied to the NGT at -40 mm Hg. The water was allowed to aspirate until a point at which either the stomach was empty or fluid ceased to aspirate out (and remained ceased for 5 min), suggesting a blockage. Once this point was reached, the tubes were flushed according to either the conventional technique or the new, air-circulating technique (see below), following which the trial was terminated and the volume of fluid aspirated was subtracted from the fluid instilled (the initial 500 mL + the amount flushed) to obtain GRV. Experiments were done in triplicate.
NGTs from Covidien (Mansfield, MA) were employed for all studies (“Salem Sump™ Dual Lumen Stomach Tube,” 18 Fr [6.0 mm] x 48" [122 cm]).
The conventional technique: A widely used technique was used as the conventional technique  to maintain patency of an appropriately placed NGT. The main, suction port was flushed with 30 mL of saline using an Asepto syringe, a GRV was aspirated using the syringe, and then 10 mL air were injected into the small, blue, side port, and finally, suction reconnected.
Eight Steps of the New Flushing Protocol, with Rationale
1. During flushing, change suction to continuous. Be sure that the tubing is not clogged.
So that suction does not turn off during the subsequent steps below.
2. Inject 120 mL warm tap water and 120 mL air into larger (clear, suction) port.
So that the stomach lining is pushed away from the tube, and so that there is an adequate volume in the stomach to suction out.
3. Reapply continuous suction; observe and note the character and amount output (Suctioned volumes are not re-administered to the patient).
So that the volume of water injected and gastric fluid already present is suctioned out. Sometimes, repeating step 2 is necessary to get this volume to suction out.
4. Flush 60 mL air into smaller (blue or clear, air-sump) port and watch for air to suction out larger (clear, suction) port.
If the NGT is in an empty stomach, the air injected in the small port will enter the stomach, suction out the suction port, and be seen as large bubbles of air in the suction tubing, but if the NGT is at the bottom of a pool of liquid, this will not happen because the air will simply bubble to the top of the gastric pool.
5. May repeat the 60 mL air into smaller (blue or clear, air-sump) port x3.
Sometimes more air is needed; may also repeat step 2 again here.
6. Call ordering physician if air flushed into smaller (blue or clear, air-sump) port is not suctioned out through larger (clear, suction) port.
Because this means that the stomach may be full of liquid and dangerously distended.
7. If air flushed into smaller port is seen to suction out through the larger port, then intake and output may be recorded, and GRV calculated by subtracting flush volume from total volume suctioned (all flushes should be recorded as intake, and all aspirate as output).
If air flushed into smaller port is immediately suctioned through larger port, then the stomach is empty (Fig. 4a).
8. Return suction to low intermittent suction.
Intermittent is better than continuous suction because intermittent lapses in allow the stomach lining is allowed to fall away from the suction holes of the NGT.
All trials were performed a minimum of three times. Data are presented as mean +/- standard deviation. A paired t-test was used to test for significance. Two-tailed significance was accepted at P <0.05.
Not a single study was found regarding how to ensure that a GRV detected after NGT flushing is accurate. Several studies distinguished the volume of gastric secretions/flushes from the volume of tube feeding [17–19], but none addressed how to determine that the aspirated GRV was the true GRV.
In vitro experiments
The in vitro model of the stomach, using a plastic bag to model the gastric mucosa, was initially compared to a similar model without the plastic bag, to validate the modeling of the mucosa. As expected, an NGT placed within a simple canister without the mucosa-modeling bag to interfere with suctioning, evacuated all 500 mL of fluid in each of three trials. When the mucosa-modeling bag was added, however, only 142 mL (+/- 37.6 mL) of the 500 mL was suctioned out before the mucosa-like bag was observed to suction into, and thereby block the NGT orifices (N = 6, including three conventional-flush trials and three new-flush trials), confirming the ability of the bag-in-canister model to mimic the suctioning of mucosa into the NGT orifices.
Following cessation of flow due to blocked NGT orifices (this occurred every time), two flushing techniques were compared, the conventional technique, and the new, air-circulating technique. The volume at which the aspiration ceased due to blocked NGT orifices did not significantly differ (P = 0.7) for the conventional-flush group (133 mL +/- 29 mL) and the new-flush group (150 mL +/- 50 mL), suggesting that subsequent comparisons are valid.
In vivo experiments
Following NGT flushing, air is sometimes heard suctioning in through the smaller, air-sump side port, creating a reassuring whistle. This sound demonstrates that the stomach is empty and the side port is patent as an air vent, aspirating air (whistling) from the room through the side port, into the stomach, and then out the suction-port tubing and into the wall. A caveat regarding the whistle is that it is not heard constantly. In fact, if it is heard constantly, the location of the tube should be reassessed because an NGT which has been pulled partially out, and has its tip in the distal esophagus, will sump air more constantly than an NGT correctly placed in the stomach. This occurs because the distal esophageal mucosa is less likely to suction into, and block, the NGT orifices, compared with the more redundant stomach mucosa.
This study has several limitations. Nonhuman stomachs were used and nonhuman studies are not always amenable to extrapolation to humans. However, pigs stomachs are similar to human stomachs. These pigs were not actually alive at the time of the experiment and following death tissues undergo rigor mortis, which may mitigate the usefulness of the pig model. However, rigor mortis does not begin to occur until 3–4 hours following death, and these experiments took place within the first 30 min following death.
Only when air flushed into the side port is seen to return out through the main, suctioning port may the tube be deemed properly functional, and accordingly then may GRV be assessed. This technique should be widely taught and adopted to minimize risk of NGT dysfunction and aspiration.
The authors would like to thank statistician Anne M. Sill, MSHS, for review of statistical analyses, Jenny Lazarus, MD for the artwork, and both Sue Eller of the Johns Hopkins Minimally Invasive Surgical Training and Innovation Center, and Matthew Gotwols of Covidien, for their cooperation in the use of the pig lab.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Kneuertz PJ, Cunningham SC, Cameron JL, Torrez S, Tapazoglou N, Herman JM, et al. Palliative surgical management of patients with unresectable pancreatic adenocarcinoma: trends and lessons learned from a large, single institution experience. J Gastrointest Surg. 2011;15(11):1917–27.View ArticlePubMedPubMed CentralGoogle Scholar
- Cunningham SC, Choti MA. Bellavance EC. Palliation of hepatic tumors. Surg Oncol: Pawlik TM; 2007.Google Scholar
- Cunningham SC, Schulick RD. Palliative management of gastric cancer. Surg Oncol. 2007;16(4):267–75.View ArticlePubMedGoogle Scholar
- Taylor MR, Lalani N. Adult small bowel obstruction. Acad Emerg Med. 2013;20(6):528–44.View ArticlePubMedGoogle Scholar
- DeLegge MH. Aspiration pneumonia: incidence, mortality, and at-risk populations. JPEN J Parenter Enteral Nutr. 2002;26(6 Suppl):S19–24. discussion S24-15.View ArticlePubMedGoogle Scholar
- Johnson AD. Assessing gastric residual volumes. Crit Care Nurse. 2009;29(5):72–3.View ArticlePubMedGoogle Scholar
- Makic MB, VonRueden KT, Rauen CA, Chadwick J. Evidence-based practice habits: putting more sacred cows out to pasture. Crit Care Nurse. 2011;31(2):38–61. quiz 62.View ArticlePubMedGoogle Scholar
- Williams TA, Leslie GD. A review of the nursing care of enteral feeding tubes in critically ill adults: part II. Intensive Crit Care Nurs. 2005;21(1):5–15.View ArticlePubMedGoogle Scholar
- Williams TA, Leslie GD. A review of the nursing care of enteral feeding tubes in critically ill adults: part I. Intensive Crit Care Nurs. 2004;20(6):330–43.View ArticlePubMedGoogle Scholar
- Sanko JS. Aspiration assessment and prevention in critically ill enterally fed patients: evidence-based recommendations for practice. Gastroenterol Nurs. 2004;27(6):279–85.View ArticlePubMedGoogle Scholar
- Ankrum JA, Olechowski A, Canseco JA, Greenblatt E, Roberts MP, Hanumara NC, et al. Nasogastric tube design to reduce clogging and simplify flushing. J Med Devices. 2011;5:027510–1.View ArticleGoogle Scholar
- Lahiri SK. Preventing gastric regurgitation with a ballooned nasogastric tube. Ann R Coll Surg Engl. 1987;69(3):122–6.PubMedPubMed CentralGoogle Scholar
- Guenter P. Safe practices for enteral nutrition in critically ill patients. Crit Care NursClin North Am. 2010;22(2):197–208.View ArticleGoogle Scholar
- Bankhead R, Boullata J, Brantley S, Corkins M, Guenter P, Krenitsky J, et al. Enteral nutrition practice recommendations. JPEN J Parenter Enteral Nutr. 2009;33(2):122–67.View ArticlePubMedGoogle Scholar
- Perry AG, Potter PA. Skill 34-4: Inserting and maintaining a nasogastric tube for gastric decompression. In: Ostendorf W, editor. Clinical Nursing Skills & Techniques, 7th Edition. Saint Louis, Missouri: Mosby Elsevier; 2010.Google Scholar
- NGT 501 (Nasogastric Tube 501): How to Make This Tube Really Suck [http://www.youtube.com/watch?v=VHmQdCTfIzY]. Accessed December 4, 2014.
- Chang WK, McClave SA, Lee MS, Chao YC. Monitoring bolus nasogastric tube feeding by the Brix value determination and residual volume measurement of gastric contents. JPEN J Parenter Enteral Nutr. 2004;28(2):105–12.View ArticlePubMedGoogle Scholar
- Chang WK, McClave SA, Hsieh CB, Chao YC. Gastric residual volume (GRV) and gastric contents measurement by refractometry. JPEN J Parenter Enteral Nutr. 2007;31(1):63–8.View ArticlePubMedGoogle Scholar
- Chang WK, McClave SA, Chao YC. Continuous nasogastric tube feeding: monitoring by combined use of refractometry and traditional gastric residual volumes. Clin Nutr. 2004;23(1):105–12.View ArticlePubMedGoogle Scholar
- Reignier J, Mercier E, Le Gouge A, Boulain T, Desachy A, Bellec F, et al. Effect of not monitoring residual gastric volume on risk of ventilator-associated pneumonia in adults receiving mechanical ventilation and early enteral feeding: a randomized controlled trial. JAMA. 2013;309(3):249–56.View ArticlePubMedGoogle Scholar
- Williams TA, Leslie G, Mills L, Leen T, Davies H, Hendron D, et al. Frequency of Aspirating Gastric Tubes for Patients Receiving Enteral Nutrition in the ICU: A Randomized Controlled Trial. JPEN J Parenter Enteral Nutr. 2013.Google Scholar
- Montejo JC, Minambres E, Bordeje L, Mesejo A, Acosta J, Heras A, et al. Gastric residual volume during enteral nutrition in ICU patients: the REGANE study. Intensive Care Med. 2010;36(8):1386–93.View ArticlePubMedGoogle Scholar
- Ahmad S, Le V, Kaitha S, Morton J, Ali T. Nasogastric tube feedings and gastric residual volume: a regional survey. South Med J. 2012;105(8):394–8.View ArticlePubMedGoogle Scholar
- Metheny NA, Stewart J, Nuetzel G, Oliver D, Clouse RE. Effect of feeding-tube properties on residual volume measurements in tube-fed patients. JPEN J Parenter Enteral Nutr. 2005;29(3):192–7.View ArticlePubMedGoogle Scholar
- McClave SA, Lukan JK, Stefater JA, Lowen CC, Looney SW, Matheson PJ, et al. Poor validity of residual volumes as a marker for risk of aspiration in critically ill patients. Crit Care Med. 2005;33(2):324–30.View ArticlePubMedGoogle Scholar
- McClave SA, Martindale RG, Vanek VW, McCarthy M, Roberts P, Directors ASPENBo, et al. Guidelines for the Provision and Assessment of Nutrition Support Therapy in the Adult Critically Ill Patient: Society of Critical Care Medicine (SCCM) and American Society for Parenteral and Enteral Nutrition (A.S.P.E.N.). JPEN J Parenter Enteral Nutr. 2009;33(3):277–316.View ArticlePubMedGoogle Scholar